Forever Tech Manufacturing Flange Ultrasonic Level Meter in Tamilnadu Forever Tech Flange ultrasonic level meteris based on the Time-of-Flight principle. Ultrasonic Level Meter sensor emitsultrasonic pulses, the surface of the media reflects the signal and the sensor detects it again. The Time-of-Flight of the reflected ultrasonic signal is directly proportional to the distance travelled. With the known tank geometry,the level can be calculated. Manufacturing Flange Ultrasonic Level Meter in Coimbatore.
Forever Tech Flange Ultrasonic level meter with ultrasonic level sensorsprovides continuous, non-contact and maintenance-free level measurement offluids, pastes, sludges and powdery to coarse bulk materials. The measurementis unaffected by dielectric constant, density or humidity and also unaffectedby build-up due to the self-cleaning effect of the level sensors. Manufacturing FlangeUltrasonic Level Meter in India.
1. Keep Ultrasonic Level Transmitter perpendicular to liquid.
2. The transducer should not be mounted too close to the tank wall, the bracketcan cause strong false echoes
3.Mount the transducer away from the inlet to avoid false echoes.
4. The transducer should not be mounted too close to the tank wall, the build-upon the tank wall cause false echoes.
5. As is illustrated by the figure on the below, the transducer should bemounted on the top of guide tube to prevent the false echoes from turbulenceand foam. The guide tube should come with a vent hole at top of the tube toallow the liquid vapor go out of the tube.