Forever Tech Manufacturing Multi-Channel Ultrasonic Flow Meterin Coimbatore. Forever Tech Multi-channel ultrasonic flow meter issuitable for continuously measuring flow and heat of clean and uniform liquidswithout large concentration suspended particles or gases industrialenvironment.
Support single channel and multi-channel at the same time, when one of channelis abnormal or not connected, it can automatically switch to single channel to working.Manufacturing Multi-Channel Ultrasonic Flow Meter in Tamilnadu.
Forever Tech Multi-Channel Ultrasonic FlowMeter Advantages
Forever Tech Multi-Channel Ultrasonic Flow Meterpipe segmentsensor is a measurement method that uses a flange to directly connect the pipesegment sensor with the pipeline to be measured. This sensor solves the problemof external and plug-in sensors caused by man-made or inaccurate pipelineparameters during the installation process. Errors cause the problem ofdecreased measurement accuracy, which has the characteristics of highmeasurement accuracy, good stability, and easy maintenance.
Forever Tech Multi-Channel Ultrasonic Flow MeterApplication
Forever Tech Multi-channelultrasonic flow meter could connect temperature sensor to become onecalorimeter and widely be used in Process control, Production measurement,Trade settlement.Water,Seawater,Acidsolution, Cooking oil,Gasoline,Coal oil,Diesel,Alcohol,Beer and Oil otheruniform liquid could transmitting ultrasonic waves.
Forever Tech Multi-Channel Ultrasonic FlowMeter Installation
Forever Tech Multi-channel ultrasonic flow metersection where the sensor of the pipe-segment ultrasonic flowmeter is locatedshould ensure that it is always filled with a steady flow of fluid (liquid)that does not scatter. This requires that the location of the sensor should beat the low end of the pipe. Both the instrument and the sensor installationlocation should be far away from the interference source. Forever Tech Manufacturing Multi-ChannelUltrasonic Flow Meter in India.
The interference source consists of two parts:
1. Interference sources that can causemechanical vibration of the measured fluid (liquid), such as water supplypumps, water supply motors, etc.
2. Electromagnetic interference sources that can cause instrument signal disorder,such as transformers, high-power motors, frequency conversion cabinets,high-voltage power supplies, and other electromagnetic interference sources.